I mean if you could get the 55 55 7A 31 from uart interface, after OpenIMU330BI updated the openimu330BI firmware. or you could save your OpenIMU330BI image bin file and send to us to check.
Cannot connect to OpenIMU330BI EVK
Here's a screenshot of the data from HTerm on COM14:
The last 6 Pages (58-63) are write protected when I first open the Target > Settings > Option Bytes window.
My Serial Number from the label on the OpenIMU330BI EVK is:
Here's another screenshot of the HTerm data with the data type set to Hex and the data scrolled back to the beginning. The first few bytes do indeed read 55 55 7A 31 as you can see.
your version of FW is 1.1.1, but calibrated area is broken, may be from your first flasing 1.1.1 you mentioned.
then pls follow below topic_5 steps to upload the bin to your IMU330:
reset the power, you should connect the ANS web.
I installed the EEPROM bin file which you sent to me exactly according to the instructions under section 5 of the link above, starting at location 0x08000000, with a size of 0x30000. I set the 6 Option Bytes afterward and restarted the IMU board. I could not get the IMU to connect. I repeated the procedure again. I still cannot connect to the IMU with the OpenIMU Monitor. What should I do?
I cannot get any data from the IMU over HTerm.
After establishing communication with the ST_LINK Utility, I tried to connect to each of the 5 COM ports using HTerm while the ST-LINK/V2 debugger LED is flashing alternately green and red. I could not get any data from the IMU over HTerm.
I also disconnected the IMU with the ST-LINK Utility and then again tried to connect to each of the 5 COM ports using HTerm while the ST-LINK/V2 debugger LED is constant red. I could not get any data from the IMU over HTerm.
- unlock the sectors
- upload bin file
- lock the sectors
- restart the IMU
if right steps, it will working on well IMU HW.
pls show pics of above steps, let's try again.
I performed the steps above exactly as listed. Here is a screenshot of the ST-LINK Utility after successfully uploading the bin file, locking the sectors, and restarting the IMU:
I immediately saw good data over HTerm on COM14:
However, I still cannot connect to the OpenIMU Monitor at developers.aceinna.com/devices/record-next
What next steps should I take?
@say Hi Say，Now you have flashed the right image, the unit output the z1 package via uart interface. please use chrome browser, download the "IMU Server Ready-to-Use (Windows)" and try again.
I downloaded the IMU Server Ready-To-Use (Windows) using the Chrome browser and installed it as shown above. I still cannot connect to the IMU using ANS. After opening ANS, when I run Server_win32_1.1.1, a DOS console window opens briefly as shown here:
It gives the message as shown: "Cannot configure port, something went wrong. Original message: .."
How should I correct this problem?
First, pls save(backup) whole EEPROM bin file, and send to us. we need to check deeply.
follow the steps in topic_5, address 0x08000000, size 0x30000.
Second, pls use the testing server.exe try again, if still not, pls send to us the Server_log file which in same file with server.exe
thanks a lot.
@say Hi say, if the IMU Server Ready-To-Use (Windows) could connected with the OpenIMU330BI successfully, there will be the log . The log picture you upload is Blurred, please send us a new one. BTW, could you send us a photo how you connected OpenIMU330BI with your computer. And if you could change a PC or USB port and test again?
I downloaded the EEPROM bin file and ran the server file you sent me with the log. I cannot attach them here. How do I send these files to you?
Here is a screenshot of the server log file:
send to us your saved EEPROM bin file and Server_log, by
firstname.lastname@example.org email to me
upload under the forum topic in below steps:
I create one testing program for you, and we will fix the bug in next official release.